The Proguard RR radiation attenuating gloves are intended to reduce the amount of scattered radiation exposure to the hands during fluoroscopic procedures.
Certified to ASTM 2547 and IEC 61331-1:2014 Standards
Proguard radiation attenuating gloves are the preferred choice for doctors in the following vocations: Radiology, Cath/EP Lab, Orthopedics, Endoscopy, Urological Procedures, Special Procedures, and Pain Management.
Price is for 1 box (5 pairs)
- Material: Lead Oxide (PbO) and Dry Natural Rubber
- Allergens: Latex Free and Powder Free
- Internal Slipcoat: Polymer Coated
- Sterility: Sterilized by Ethylene Oxide
- Post Processing: Polymer Rinsing
- Shelf Life: 3 Years as per ASTM 7160-05
- Glove Design: Hand Specific
- Cuff: Non-beaded
- Glove Length: 280mm (minimum)
- Packaging: Pouch (1 Pair); Box (5 Pairs)
|RR1||0.05mm Pb @ 60kVp||+/- 0.24mm||50%||42%||37%||33%|
|RR2||0.07mm Pb @ 60kVp||+/- 0.28mm||58%||50%||45%||41%|
|RR3||0.09mm Pb @ 60kVp||+/- 0.38mm||73%||65%||57%||52%|
Compliance Requirements: ASTM D5151, ASTM D6124, ASTM D412, ASTM F2547, EN ISO 374-1:2016, EN ISO 374-5:2016, EN 421:2010 (excl. clause 4.3) EN 420:2003 + A1:2009
Biocompatibility Compliance: ISO 10993, ASTM F 719 and ASTM F 720
Disposal: The Proguard RR glove contains lead and should be disposed of through hazardous waste procedures in accordance with local laws and regulations. The material should not be disposed of by incineration and use of self-contained breathing apparatus is recommended if the sheeting is ignited by fire.
Expiration: This is a sterilized product and expiration is set at 3 years from manufactured date per FDA requirement. his Indication may be found on the label on the original external packaging and indicates the period of time during which the device, sterilized with Ethylene Oxide, may be used as long as it has been stored properly in suitable conditions and has intact packaging.